Removable Clip System Introduction
Step 1 Insert the Polypectomy Snare and make it near the Clip
1.1, Revolve the endoscope or proper air admission , in order to put the clip in the middle of the field of vision;
1.2, Insert the Polypectomy Snare until it appears in the field of vision;
1.3, Suggest to use rotatable 10mm(width of Ele Ctrode) size snare. The wire OD should be 0.36~0.38mm.
Step 2 Adjust the Snare angle and position
2.1, Revolve the Snare or the Endoscope, make the angle between Clip and
Snare to be 60°~90°;
2.2, Keep Snare and Clip groove Not vertical .If they are already vertical, use the Endoscope left-right button to adjust the angle.
Step 3 Circle the clip
3.1, Use the Snare to circle the clip ,and then tighten the Snare.
Suggest to put the snare at clip 1/3 length position (A position).But must make sure the snare under
3.2, Always keep the clip under Snare Circle.
Step 4 Let the Snare move Clip Groove
4.1, Revolve the Endoscope ,drive the Snare completely move into the
4.2, Revolve the Endoscope to make clip left-right move ,in order to check if the
Snare already at the right position.
Step 5 Tighten the Snare
5.1, Pull the Snare handle to the end (Or till you can not pull);
5.2, Properly take back the Endoscope or put forward the Snare ,which offers a better visual field for observing the contact position between Clip and Tissue.
Step 6 Check if the Clip already drop from Tissue
6.1, Use the Endoscope to move up the Snare;
6.2, Feel the Endoscope move force to check if the already drop from Tissue;
6.3, Observe the contact position of the Clip and Tissue to check if the Clip already drop from Tissue.
Step 7 Clip take out
7.1, Take out the Endoscope which also takes the clip out of human body;
7.2, Or you can release the Snare and continue surgery ,after finish all other procedures ,pls don’t forget to take out the Clip.